Laboratory Protocol for the Seeger Passive Bar
Screw the OT Equator of the correct cuff length into the NanoDent implant analogue.
Place the burn-out castable red ring, followed by Laboratory Seeger (thin part down) tighten the screw cap in titanium without forcing.
Wax up the bar or use resin to design the passive bar incorporating castable elements.
Completed metal passive bar.
The retentive bar components can be design and added as required according to preference.
Dimensions of burn out OT equator castable used to increase bar retention and deliver parallelism.
The Seeger System has been designed to allow the construction of bars on implants even if non-parallel, with passive connection. The peek ring connects passively the bar to the OT Equator attachments to prevent unscrewing and can be removed with the titanium self-extracting screw.